Structure of the PC measurement electronics with Spider8. B-1 .. When exchanging any modules, only original HBM parts must be used. Qualified personnel. Spider8 – Easy and reliable PC-based The Spider8 is a multi-channel PC measurement electronics for parallel, dynamic . Our factory in Darmstadt or HBM. Catman talks to Spider8 remotely. Keyword: Wireless. Remote communication with field equipment up to 5km away is now possible with HBM’s improved.
|Published (Last):||1 April 2011|
|PDF File Size:||19.4 Mb|
|ePub File Size:||12.92 Mb|
|Price:||Free* [*Free Regsitration Required]|
W henever this is not true, the Spider should be turned off and on.
Lot 753: HBM Spider 8 Data Acquisition Panels
So in this example, the timeout has to be adjusted in respect to the measuring rate. Also, which drivers are you using exactly? Furthermore, the system is very unstable. This Tutorial has information regarding how to properly call functions from DLLs using the call library node.
I have run the firmware sp8firm32 and de archive sp32, both can be downloaded in the hbm website, and my example started to run.
Message 2 of With this configuration I copied the spider If you need the archives email-me wanderleyff gmail. The Spider is connected with the notebook spiver8 the serial port.
I tried to run one of the spider8 examples with labview and did not work.
I have updated everything. In most cases the baud rate will be the limiting factor, but don’t forget the measuring rate!
Soon I will tell what were the problems. Message 4 of Perhaps the old firmware caused some problems After some errors had occured, I didn’t turn the Spider off and on, and the error messages were strange. Wpider8 am reading your inputs with respect the use of the Spider8 with Labview. First of all, my operating system is NT4. Auto-suggest helps you quickly narrow down your search results by suggesting splder8 matches as you type.
It seems as the connection is not fast enough. I have read that minimum version shoud be P20 to work correcrtly with Labview. The big advantage is, that you have all options to control your Spidee8 process.
Software and Firmware Downloads for Legacy Products | HBM
A further point are inconsistencies between the online help, the help function in the Spider setup program and the printed manual! Message spideer8 of Message 10 of A big problem were the error messages which sometimes didn’t make sense to me. For Labview there are some examples which show how to access data splder8 catman activeX. I have 2 examples and jbm shows the mesured value in the port and the other plots graph in the screen of the signal mesured in the port.
May be, he has a demo version for you almost catman professional including the catman activeX interface. The cable has got a length of about 1 meter. Please could you help me to know what the correct steps are to try the spider89 works with Labview?. I would like to receive help with respect this issue. Now I have got problems with data acquisition, it doesn’t work at all.
This is not very useful! I hope this help you. Are you getting any error messages from the VI’s or does the spidsr8 just not seem to respond at all?
Any advice is welcome as well as was a place where I still can download the last firmware version. I haven’t downloaded the latest firmware yet. Last but not least the timeout has to be adjusted according to the specific number of counts, baud rate and measuring rate.